ASM Failure Analysis Case Histories: Medical and Biomedical Devices
A narrow bone plate made of type 316 stainless steel and used to stabilize an open midshaft femur fracture failed. A crack at a plate hole next to the fracture site had been revealed by a radiograph taken 13 weeks after the operation. The plate was revealed to be slightly bent in the horizontal plane, and the fracture gap was considerably open. The screws and plates supplied by different manufacturers were revealed to be different with respect to microcleanliness (primary inclusion content) of the materials and only one of them was found to be according to specifications. The local crack formation was influenced by the presence of larger inclusions. The screw failed was revealed to have failed through a fatigue mechanism by the presence of striations in the scanning electron micrograph. The crack in the plate was revealed to have originated at the upper, outer corner of the plate by the beach marks which indicated the action of asymmetric bending and rotational forces.
Bone Plates, ASM Failure Analysis Case Histories: Medical and Biomedical Devices, ASM International, 2019
Download citation file:
New Handbook: Volume 11B
Volume 11B serves as a reference and guide to help engineers determine the causes of failure in plastic components and make corrective adjustments through design and manufacturing modifications.